A series of experiments has been carried out in order to characterize a miniannular phased array applicator prior to possible clinical implementation. The energy deposition patterns over the frequency range of 100 to 200 MHz were determined in several human limb models of different complexities by measuring the electric field strength patterns. The point of maximum energy deposition within a homogeneous, muscle-equivalent cylindrical phantom positioned coaxially within the MAPA was found to be at the center of the applicator. The energy deposition patterns seem to be more uniform at the lower frequencies. Inclusion of a cylindrical bone-equivalent phantom positioned coaxially with this muscle-equivalent phantom does not seem to significantly alter the energy deposition patterns in the muscle-equivalent region. For more realistically shaped, homogeneous muscle-equivalent limb models, the resulting energy deposition patterns appear to be confined mostly to the intended treatment region. However, the point of maximum energy deposition was not at the middle of the applicator as with the cylindrical model, but shifted towards a smaller cross-sectional region. This shift in location of the point of maximum energy deposition varies with the location of the MAPA on the limb. A secondary region of high-field strength was also observed at the ankle for a MAPA centered about the knee. In this study, the energy deposition patterns appear to be significantly dependent on the shape of the model. Therefore, this factor must be taken into consideration for the proper prediction and control of the heating patterns resulting from the use of this type of applicator for clinical hyperthermia treatment.