This experiment was designed to study if liquid conservation buffer used in the novel Tissue Sampling Technology (TST) from Allflex can be used for Illumina BeadChip genotyping. Ear punches were collected from 6 bovine samples, using both the Tissue Sampling Unit (TSU) as well as the Total Tagger Universal (TTU) collection system. The stability of the liquid conservation buffer was tested by genotyping samples on Illumina BeadChips, incubated at 0, 3, 15, 24, 48, 72, 168, 336, 720 h after sample collection. Additionally, a replenishment study was designed to test how often the liquid conservation buffer could be completely replenished before a significant call rate drop could be observed. Results from the stability study showed an average call rate of 0.993 for samples collected with the TSU system and 0.953 for samples collected with the TTU system, both exceeding the inclusion threshold call rate of 0.85. As an additional control, the identity of the individual animals was confirmed using the International Society of Animal Genetics (ISAG) recommended SNP panel. The replenishment study revealed a slight drop in the sample call rate after replenishing the conservation buffer for the fourth time for the TSU as well as the TTU samples. In routine analysis, this application allows for multiple experiments to be performed on the liquid conservation buffer, while maintaining the tissue samples for future use. The data collected in this study shows that the liquid conservation buffer used in the TST system can be used for Illumina BeadChip genotyping applications.