Dedicated paclitaxel-eluting bifurcation stent BiOSS® (bifurcation optimisation stent system): 12-month results from a prospective registry of consecutive all-comers population.
The aim of this study was to analyze the difference in neointima pattern assessed by intravascular ultrasound (IVUS) between two dedicated bifurcation stents, BiOSS® Expert and BiOSS® LIM at 12-month follow-up. This manuscript reports IVUS findings obtained from the analysis of patients enrolled into first-in-man registries initially assessing the BiOSS Expert® (paclitaxel) and BiOSS LIM® (sirolimus) stents. Quantitative angiographic analysis was performed pre, post-stenting, and at follow-up. IVUS examination was performed at 12 months. There were analyzed 34 cases (BiOSS Expert® 11 patients, BiOSS LIM® 23 patients). Procedural characteristics in the two groups were similar, except for rates of main vessel predilatation and FKB/POT, which were higher in BiOSS® LIM group, 54.5 % vs 73.9 % (P < 0.05) and 0 % vs 39.1 % (P < 0.05), respectively. When comparing late lumen loss (LLL) for both stents there were significantly bigger values for main vessel and main branch in the BiOSS® Expert group, but not in side branch. Intravascular ultrasound examination showed that in the BiOSS LIM® group comparing with the BiOSS Expert® group there was lower neointima burden in the whole stent (24.7 ± 7.5 % vs 19.4 ± 8.6 %, P < 0.05) as well as in main vessel (22.8 ± 5.6 % vs 16.9 ± 6.1 %, P < 0.05) and main branch (36.1 ± 6.5 % vs 27.6 ± 8.7 %, P < 0.05), but not at the level of bifurcation (15.1 ± 3.8 % vs 13.6 ± 5.4 %, P = NS). In addition, we found that final kissing balloon/proximal optimization technique (FKB/POT) was associated with significantly smaller value of LLL in main vessel (0.24 ± 0.09 mm vs 0.32 ± 0.14 mm, P < 0.05), which in IVUS analysis resulted in smaller neointima burden in main vessel (13.7 ± 3.9 % vs 18.9 ± 4.45 %, P < 0.05) as well as at the bifurcation site (12.6 ± 4.1 % vs 14.1 ± 2.4 %, P < 0.05). The obtained results suggest that neointima proliferation was the largest in main branches of both stents assessed in quantitative angiography (LLL) as well as in IVUS (neointima burden) and the neointima increase was smaller in BiOSS LIM® stents than in BiOSS Expert® stents. Moreover, the middle part of the stent seems to not to be associated with excessive neointima proliferation and more aggressive protocol of implantation with the use FKB/POT seems to decrease this process.